The Elcometer range of digital coating thickness gauges has been specifically designed to provide highly accurate, reliable and repeatable coating thickness. The Elcometer Coating Thickness. Gauge is an incredibly versatile gauge. With a range of probes in both Integral or Separate probe versions for coating. Elcometer Coating Thickness Gauge. Operating Instructions. English. R. English Equipment described in these instructions is covered by the following.
|Published (Last):||7 October 2009|
|PDF File Size:||11.71 Mb|
|ePub File Size:||5.82 Mb|
|Price:||Free* [*Free Regsitration Required]|
Gauges with an integral probe will switch on automatically by placing the probe against a hard surface. Do not operate the Elcometer without a probe connected.
Your gauge can display statistical summary information in a preset sequence. To maximise the benefits of your new Elcometer Coating Thickness Eldometer please take some time to read these Operating Instructions.
Elcometer Unit is in great shape and has been tested against NIST calibration standards and holds calibration great. Repeat for all statistics wanted. Consult your local Environmental Authority for further guidance.
Elcometer 345 Manual
Elcometer User Manual for Mac comes with powerful features such as editing Documents. Learn More – opens in a slcometer window or tab Any international postage is paid in part to Pitney Elcoemter Inc.
The principle of electromagnetic induction is used for non-magnetic coatings on magnetic substrates such as steel. The gauge is available either with a built-in integral probe or with a separate probe. To extend probe life, always set the probe down so that it is perpendicular to the panel surface.
In this routine if, after the key-lock feature is operated, 7 seconds elapses without pressing a key or rlcometer a reading, the gauge will automatically continue to the next step, i. If the display is heated above 50C F it may be damaged.
There are many Updated: Immediate payment required for this item. Users of the Elcometer may also benefit from the following Elcometer products: The New Elcometer Automotive Paint Meter The next generation automotive refinishing gauge for instant measurement of paint thickness.
Elcometer Coating Thickness Gauge
For further details see EN ‘Paints and Varnishes – Corrosion protection of steel structures by protective paint systems – Measurement and acceptance criteria elcomteer the dry film thickness. Instrucciones de Operacin Elcometer Documents. Correct printer fault or press to cancel print function. Place the chosen high calibration foil on the bare metal surface and take a number of readings until a stable average is displayed.
Elcometer NB Coating Thickness Gauge | eBay
See To Recall The Factory Calibration on page 20 for a method to deliberately reset the factory calibration setting. V Adaptor for pipes: No additional import charges on delivery. Please enter up to 7 characters for the postcode. Keep pressing Keep pressing Keep pressing Keep pressing Press 24 until until until until1Fis displayed, pressto accept to accept to accept to accept2FLois displayed, press is displayed, press is displayed, press3F4F. Comes with wired probe, and calibration shims not pictured.
Press and release lecometer begin the special calibration sequence. If or is released within 4 seconds the gauge will display the value for a further 3 seconds during which time or can be pressed again.
Calibration adjustment should be carried out with the appropriate probe, on the elcometeer type of metal, the same curvature and similar surface finish as that of the item to be measured. Details of Elcometer offices around the world are given on the outside cover of these Operating Instructions.
The bleeper will sound to indicate acceptance of the new calibration. Remove the elcomfter from the gauge if it is to remain unused for a long period of time. Whichever is the greater Separate probe versions only Whichever is the greater when calibrated close to the thickness to be measured.